|Reference Standards||Standards the instrument complies with: ISO 6916-2; ISO 293; ISO 2393; ASTM D3182; ISO 6916-1; UNI 5572;|
|Max useable platen surface 250 mm x 250 mm||Max closure force 25 Tons||Max platen stroke 200 mm||Temperatures
Up to 250°C (300°C optional).
Temperature uniformity in the area 200x200mm:
Maximimum deviation from the centre of the platen <0,5°C.
Mean difference of platen temperature from set point <0,5°C.
|Cooling System (Option) Cooling of the platens with controlled cooling gradient. The cooling is controlled using the Automatic Press control PLC device||Closure force Control(option) Continuous control of the closing force with regulation between 10 and 250 kN with 10N regulation steps. Max. regulation error: 1 kN.|
|Press control device|
|Touch Screen PLC controller PLC with touch-screen display. Permits to: • set displacements of the moving plate • set the temperatures • activate the cooling and cooling rate (with cooling option) • set closure force (with force regulation option)|
• Safety Pushbutton.
• Doors with safety switch
|Labelling CE Labelling|
|Calibration Report Calibration report with traceability to primary standards in conformity with the Calibration requirements specified in ISO 2393 standard||Calibrated parameters
The certificate includes the calibration of:
- temperature of the platens. The measure is performed in 9 different points of the platens and at 3 different temperatures (140°C, 170°C, 200°C).
- Closure force (with Pressure control option)
|Platen temperature control
- Electrical heating.
- Independent temperature control of the plates by means of PID controllers
|Power supply 380 VAC ±10%,50 Hz ±3, 10 A,three-phase -Other on request||Electrical power 6 KW||Instrument dimensions (with mounted side drawer) (Width x Depth x Height) 1200 x 950 x 1800 mm||Instrument dimensions (without side drawer) (Width x Depth x Height) 650 x 950 x 1800 mm||Weight 520 Kg|
|Cooling System The cooling is perform with the connection too a water source (or a chiller). The control of the cooling and of the cooling rate are controlled by the PLC controller.||Connection to cooling liquid
The water supplied to the Press must be demineralized.
The fittings for the inlet water is 1/4"
|Insulated tank for cooling water (optional) The hot water, which exits from the cooling circuit, must be cooled before being discharged. The insulated tank permits to store and cool the water before the discharge||Water consumption The cooling of the platens of the press, starting from 250°C down to 70°C requires approxiametely 30 liters of water. The quantity changes depending from the temperature of the water used.|
|Gibitre Press Logger
Software compatible with Windows 7, 8 and 10 for continuous recording of the press conditions (temperature, pressure, moving platen position, cycle step).
The software is connected to the press with USB cable